Start by switching on the twilite system with the ON/OFF button. After the initialization procedure has completed, press the green ‘play’ button on the touch screen to switch on the PMTs.
Start PMOD on the acquisition computer and open the acquisition window with the Twilite Acquisition button in the PMOD ToolBox.
It is assumed that connectivity with the twilite as been configured appropriately.
In a simultaneous experiment with a PET acquisition please make sure that the system clocks of the acquisition computer and the PET system are synchronized before starting PMOD (PSAMPLE time stamps each data point, using the system time as reference). Furthermore, in order to acquire enough background data for automatic background subtraction during AIF correction, start the twilite at least 20 sec before tracer injection.